qsUtility Data Analyzer Vpro
SDDS FILE: /cs/prohome/apps/q/qsUtility/pro/fileio/data/IHAM601/IHAM601_x_zigzag_2020-08-18_13:29:03/IHAM601_zigzag_2020-08-18_13:29:03_qsUtility_analyzer.sdds
CED WORKSPACE: Optics CED LOGIN KEY: ued
Nice sleuthing. Both Wiens are between 1I02 and 0I02, so four power supplies for the four electrodes, maybe explains similar but separate ~10 kHz ferquencies. Note, we're replacing all those supplies this SAD. An option for further troubleshooting is to set WIens for 0 deg, turn the HVPS OFF, and re-measure beam motion.