IHA0I02 - control cables terminated at rack IN03B07 and IN01B03. Anti-collision circuit cables terminated at device location. Cross connect jumpers configured in rack IN03B07. Still need to reconfigure the cross connect jumpers in rack IN01B03. We plan on testing the interlock / anti-collision circuit tomorrow morning.
While installing the new collision circuit and troubleshooting
ITV0I04 and the Master slit, the out limit switch for the Master
slit that connects to the collision circuit was found to be bad,
also the out limit switch for the viewer had to be readjusted,
along with replacing a bad MAC valve that was not driving in
the viewer when actuated. ITV0I04 and Master slit are