New VME cards created and merged in CED:
FSD_IN03_2, BLMIN01, BLMIN02, BLMIN03,
Old CAMAC cards removed from CED:
FSD_IN03_3 and FSD_IN03_4.
Old CAMAC FSD_IN03_2 was deleted and replaced with new VME
FSD_IN03_2 in CED.
Physical VME, FSD and BLM card installation and cable
re-routing, repair, and update is in progress. Software
implementation is also in progress..
Fixed the 1A31 Muxed 4ch stack today. Originally it looked like the
Mux Box was bad however it turned out to be the RF module. The
stack passes Local Alpha's, Supercal and Global Supercal. After
fixing the 1A31 stack I ran Global Supercal on 4ch bpms for
IOCEA1, IOCEA2, IOCEA3, IOCEA4, IOCEA5, IOCEA6. The
bpms on those IOC's all passed and I saved their IOC's. IOCWA1
is temporarily powered down currently so its 4ch bpm has not
been Supercal'd or had Global Supercal performed on it. Maybe
next week when power is back.
Heat Stress survey results were below the occupational
exposure limit. A work/rest regimen is not required.
Magnet surface temperatures measured during the survey are
within recommended temperatures for unprotected metallic
surfaces. No further action is required to protect staff
from incidental contact.